On this page:
|By: elsonds added:27-02-2014 16:08 (modified: 15-05-2014 15:13)|
Information on research areas and activities can be found on my research website:
List of individual projects and links to websites:
- Smart sensing for surgery, funder EPSRC EP/L014149/1
- Robotic assisted surgical guidance and visualisation, funder DTI/TSB/EPSRC
- Realtime in vivo in situ surgical sensing and imaging, funder TSB/EPSRC
- Multi-modal optical imaging applied to robotic assisted endoscopy, funder EPSRC
- Endoscopic FLIM for label-free tissue contrast, funder EPSRC
- Blue laser diode endo-illuminator, funder NEAT/DoH
- Development of miniature wireless GRIN lens video endoscope, funder NEAT/DoH
- PAtient Specific Simulation and PreOperative Realistic Training for liver Surgery (PASSPORT), EU FP7
- OPTIMISE Optical Platform for Therapy and Diagnostic Imaging in Minimally Invasive Surgical Endoscopy, ERC StG
Ultrasound mediated tomography and elastography
|Tracking shear waves in turbid medium by light: theory, simulation and experiment|
Experimental set-up showing (a) shear wave near surface, (b) shear wave at depth.
Simulated and experimental shear wave propagation results.
|By: elsonds added:12-03-2014 11:23 (modified: 15-05-2014 15:15)|
|Shear wave elasticity imaging based on acoustic radiation force and optical detection|
Experiment set-up. DAO: the distance between optical and acoustic axis, which can be adjusted by using the translation stage (not shown in the figure), AMP: power amplifier, WG: waveform generator and DG: delay generator. The ultrasound absorbing material minimises the ultrasound reflection. The size of the tank is 30 3 20 3 7.8 cm.
Scanning scheme for the heterogeneous phantom. Position O indicates the starting position for optical axis in two scanning runs. A and B are two different acoustic axis positions.
Optical contrast difference (DC) as a function of time and DAO in three homogenous phantoms of different stiffness. It can be seen that for a given DAO, the optical signal always peaks earlier and is better defined for a stiffer phantom.
|By: elsonds added:15-05-2014 15:13 (modified: 15-05-2014 15:15)|
|Viscosity measurement based on shear-wave laser speckle contrast analysis|
Shear-wave laser speckle contrast analysis (SW-LASCA) experimental setup
Schematic for detecting continuous shear waves with SWLASCA method.
Speckle contrast difference due to continuous shear wave generated by an ultrasound mechanical index (MI) of 1.28. Standard deviation is produced with three repeated measurements.
Dr Lei Su, Queen Mary University of London, Single fibre endoscopy, 2016
Professor Hui MA, Tsinghua University, China, Polarization-resolved imaging, 2016
Lipei Song, Institute of Modern Optics, Nankai University, Tianjin, 300071, China, Laser Speckle Contrast Imaging, 2013
Lena Maier-Hein, German Cancer Research Centre (DKFZ), 2011
Danail Stoyanov, University College London, 2008
Professor Laura Marcu, University of California, Davis, Time-resolved fluorescence spectroscopy, 2006
Research Student Supervision
Avila,F, Concurrent Robotic Red-Flagging Framework for GI endoscopy
KOH,KEVIN, Multimodal Multispectral Optical Endoscopic Imaging with Endomicroscopy for Biomedical Applications
Li,R, Ultrasound modulated optical tomography in optical diffuse medium using acoustic radiation force
Lin,J, A self-calibrated endoscopic structured light system using multispectral detection
Qi,J, Development of Optical Polarisation Resolved Endoscopy
Sauvage,V, Fluorescence and Diffuse Reflectance Spectroscopy and Endoscopy for Tissue Analysis
Song,L, Endoscopic Laser Speckle Contrast Analysis for Tissue Perfusion
WOOD,TOBIAS, Dynamic Hyperspectral and Polarized Endoscopic Imaging
Zhou,Y, The Applications of Gold Nanoparticles in Photothermal Therapy for Cancer Treatment